D CV. Read this owner’s manual thoroughly before use Owner’s operation manual for series. DT Series. 3 1/2 Digital Multimeter. Read this owner’s manual thoroughly before use. DT series multimeter have been designed according to IEC concerning electronic measuring. View and Download Histest DT owner’s operation manual online. DT Multimeter pdf manual download.
|Published (Last):||15 July 2017|
|PDF File Size:||8.27 Mb|
|ePub File Size:||10.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set the range switch to the desired range.
Mnuale case and tighten screws completely before using the meter to avoid electrical shock hazard. Table of Contents Add to my manuals Add. To print the manual completely, please, download it. Operating Instruction “COM” jack.
The voltage is approximate 5Vp-p. Before measuring current, check the meter’s fuses and turn power off to the circuit before connecting the meter to the circuit.
Set the range switch to ” ” or dt830v ” range. Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
Safety Information If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. If the resistor to be measured is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance.
Front Panel Description This function allows the meter to output a signal as a signal generator.
DTD Manual Ranging Digital Multimeter
This function allows the meter to output a signal as a signal generator. Use the proper terminals, function, and range for your measurements. Page of 13 Go. A test signal will appear between the and “COM” jacks. To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes.
Don’t show me this message again. Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance.
Histest DT830 Owner’s Operation Manual
Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected manualf measurement circuits. In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user.
A coupling capacitor should be used when connecting the meter to a circuit. Enter text from picture: Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes. By pressing ‘print’ button you will print only current page.
Got it, continue to print. If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost. Comments to this Manuals Your Name. Don’t have an account? Related Manuals for histest DT No related manuals.